Don’t have an account? Scale will display “”, when power is turned on and weight is present. IF you are using a remote display, check the main display first, if it’s OK then check the cable and plug connection on the remote First unplug, then plug in the AC adapter to reset the program. If the corners are not within one 1 division of each other, conduct a good visual inspection to verify that there are no binds or interference with the platform and re test the scale. Using the level bubble, adjust one or two feet minimally to level the platform. Will usually ship within 3 business days of receiving cleared payment – opens in a new window or tab. Page 16 The scale must be calibrated if you proceed past this step.
|Date Added:||5 October 2006|
|File Size:||25.21 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Calibration Procedures -8, -7, -6, -5, -4, -3, -2, -1, 0, 1, 2, 3, 4, 5, 6, 7, 8, 9, 10 The breakdown of the zones are as follows: To assist with customer satisfaction, we gather and track all fakrbanks by machine type, model, serial number, page count for printers and IMEI number for telecommunication items as part of the process, prior to shipping the product to ensure order accuracy.
Image not available Photos not available for this variation.
Sales tax may apply when shipping to: Page 15 Section 6: Seller assumes all responsibility for this listing. People who viewed this item also viewed.
Fairbanks Ultegra SCB-R9000-14U Service Manual
Ratings and Reviews Write a review. Contact the seller – opens in a new window or tab and request a shipping method to your location. This manual also for: Learn more – opens in a new window or tab. Back to home page. Specifications Page 13 – Environment Page 14 – Section 6: Repair and Parts Replacement Buyer pays return shipping.
The scale must be calibrated if you proceed past this step.
We appreciate your business! Turn the unit upside down and remove all four feet from the scale platform 3.
If the corners are still not within one 1 division of each other, identify the fairnanks s with the error and replace the load cell in that corner. Our business working hours are 7: Please enter a number less than or equal to Minimum monthly payments are required.
It uses default Windows drivers and is installed automatically fairbahks first connection with a PC. Using the level bubble, adjust one or two feet minimally to level the platform.
By pressing ‘print’ button you will print only current page. All equipment should be protected from direct sunlight.
Thanks for the assistance Great value, works great and made it here to save me the headache of shipping errors X Previous image. Display Assembly Replacement Section 7: Don’t have an account?
Scale instrument, modular designed instrument, configurable and upgradable using printed circuit modules pages.
Fairbanks Ultegra Scb-ru Digital Scale lb Capacity | eBay
Scale will display “”, when power is turned on and weight is present. For additional information, see the Global Shipping Program terms and conditions fairban,s opens in a new window or tab. No additional import charges at delivery! Screw in all 4 feet, set upright and re-level See all condition definitions – opens in a new window or tab.
We can combine shipping on purchases of certain items that can be combined into the same box.
Zone Latitude degrees Reference City